Digestion of Secondary Recovered Fuel Samples with PDC
The Multiwave 7101/7301/7501 series with its PDC (Pressurized Digestion Cavity) offers the opportunity to digest any of the diverse fractions of secondary recovered fuels for subsequent element analysis even within one run.
Secondary Recovered Fuels (SRF) are different streams of material from the garbage recycling and are a challenging mix of materials. In many countries there are limits of heavy metals for this type of fuel, as they are emitted into the environment after the incineration.
As secondary recovered fuels consist of very different materials like polymers, wood chips, ground used tires etc., the chemical behavior of the material can vary considerably due to its origin, main components, but also fillers. This also makes the homogeneity of the sample a major issue. Moreover, as it is not predictable, which types of polymers are present in the secondary fuel samples, the method should be tolerant and be capable to handle the mentioned variability without deviation in recoveries.
For a successful acid digestion a sufficiently high temperature has to be provided to be able to digest the more durable polymer components like aromatic structures. However, fillers like SiO2 or pigments like TiO2 will only dissolve with the presence of HF in the acid mixture. The recovery of other elements like Fe, Cd benefits from the presence of the complexing hydrochloric acid.
In order to demonstrate the suitability of Multiwave 7101/7301/7501 series for sample preparation prior to element analysis the recovery rates of an exemplary certified reference material were determined and in addition a real-life sample was compared with a validated reference digestion with a Multiwave 3000. The described conditions may be directly applicable to secondary recovered fuels or can be used as a starting point for a sample specific optimization.
Get the document
To receive this document please enter your email below.